Welcome to our site! Electro Tech is an online community (with over 170,000 members) who enjoy talking about and building electronic circuits, projects and gadgets. To participate you need to register. Registration is free. Click here to register now.
anybody know what the problem is with a gould 400 scope that i have.
on the screen is a message saying calibration in progress...
if i press the all button a series of horizontal lines appear for a couple of seconds, thats all, nothing else.
any help appreciated.
hi. i replaced 9v regulator, this is under the crt where its flared out, near to R35, this NEARLY fixed it, lines and cal message gone, screen back to its more recognisable disply, however did not display applied known good waveform.
the regulator before replacement was getting hot very quickly, it still got hot after replacement, it lasted about 15 mins then failed, must be something downstream from this regulator,prob. a duff capacitor. see if your regulator gets very hot when on. i have put it to one side , got a bit fed up with it, however i will have another shot at trying to trace the faults, any success and i will post results.
Try checking the capacitors UPstream, they are likely to be in a PSU section that might be a switchmode PSU in a metal cage etc. It's probably high freq ripple that is being fed to the input of your 9v regulator.
Now if you had a scope you could look at the waveform of the PSU rails...