Electro Tech is an online community (with over 170,000 members) who enjoy talking about and building electronic circuits, projects and gadgets. To participate you need to register. Registration is free. Click here to register now.
Welcome to our site! Electro Tech is an online community (with over 170,000 members) who enjoy talking about and building electronic circuits, projects and gadgets. To participate you need to register. Registration is free. Click here to register now.
This has two variable driven capacitors,one vacuum one open air.the vacuum has a 10 turn pot for reference / feedback. It snaps the stop in the pot then
Damage inside of the pot.we have set it for calibrated position.we have sent other the match and controller to match back to factory nothing wrong with controller and they set up the match network.
I have google this model and sent request for information to 5 university that have the same unit.no replies yet.
Thanks for the reply
This is from the other person that operates it and also repairs on it..
I'm able to tune some of the recipes manually. It struggles or won't tune at all, even in manual, on other recipes. We don't have a schematic
This week I haven't had the time .I only work 2 days week.I need to verify all the dc voltages and check 4 any ripple.
The info we do have is just breakdown of parts and interconnect wiring.also the opeating manual.
Two large files.
"not tuning" might actually be "no plasma", if someone else is reporting the problem. That can usually be remedied by increasing the pressure until you get ignition. That was common on a manual sputtering system. It didn't happen to often, but it did.
There was a plasma view port. This was manual tune only. It also did DC sputtering.
On an automated PCVD system I helped put together, there was a Plasma Bias output on the ENI RF power supply. I interfaced that using an Analog Devices isolation amplifier, so we could determine ignition and get the correct deposition times. I THINK it mattered where you set the initial conditions too. I didn't do the recipies, but I did do all of the electrical, RF and electronics.