i was taking measures with ESR meter in case of bad caps. then i saw that it broke, don't know why, but it shows .25 value instantly upon startup. no shorts, bad solderings, cracks etc to be seen. and, i discharged the cap i working on, and this meter have over-voltage protection. well, i contacted the manufacturer, and he agreed to repair it for no cost. I must ask him to tell what was the fault, if he repairs it, as it would be excellent education for me too. well, **** happens sometimes, luckily it wasn't scope (knockin' on wood here...)
i havent sent it yet, so if you have any ideas, feel free to tell them the model is EVB ESR meter
Blog entry posted in 'Other, ''random''', Jul 9, 2013.